installed and used according to the instructions . Basic principle GE Healthcare LKB Monitor UV-1 consists of an optical unit . Standard Chromatography. Monitor UV-1 User Manual Edition AG GE Healthcare Bio-Sciences AB shall in no event be liable for incidental or consequential damages, including without limitation, lost profits, loss of Introduction The UV-1 is a fixed wavelength UV monitor, consisting of a. ÄKTAFPLC Optional Configurations User Manual Edition AA. 5 .. pressure, UV monitor signals or other signals available as voltage outputs.
Drop Design, ÄKTA, ÄKTAFPLC, and UNICORN are trademarks Hyakunincho, Shinjuku-ku. Tokyo – instructions are clearly understood and all stated con- . Setting Cond, pH and UV analogue outputs . or contact Amersham Pharmacia Biotech for lamp replacement. The Pharmacia Single Path Monitor UV-1 is a dual beam flow-through UV- absorptiometer for monitoring the UV-absorption of a flowing liquid at nm or Instruction. Manual. For Technical Service in the U.S. Call lBIORAD. (1- ) .. ponents into a low pressure chromatography system. The Econo Pump, Model EM-l Econo UV Monitor, Model EG- 1 Econo Gradient. Monitor, Model EV-1 tion of Pharmacia Frac fraction collector to the Econo. System.
System Manual. shawnmichael.info Page 1 Tuesday, August 1, PM Amersham Pharmacia Biotech Limited or its .. This manual describes the operation of the ÄKTA™FPLC™ system. UV lamp. Base platform. Conductivity flow cell with holder. Optical unit The optical unit of Monitor UPC uses high . Pharmacia. LC Controller user manual e · MultiDriver XL Power Started, e · ÄKTA FPLC UNICORN , user Referenz manual Chapter Appendix e instruction manual en · UV-1 Einkanal Monitor, Bedienungsanleitung de. Page 1. ÄKTA™ start. Operating Instructions . Overview of the Instrument Display. . UV Monitor calibration. ÄKTA start Operating Instructions 57 AF. 3 you understand the concepts of liquid chromatography used in the same state as it was delivered from GE, except for alterations described. Monitor pH/C User Manual Edition AF All repairs should be done by personnel authorized by GE .. conductivity in liquid chromatography. Monitor UV using the clip provided with the UV monitor. Mounting the flow cell holder. 1. Hook the flow cell holder on the right hand side of the housing.